How often do I need my X-ray equipment inspecting?
Most manufacturers recommend you have an annual maintenance check.
For radiological testing, a test should be undertaken at least once every three years for intra-oral, panoramic, panoramic/cephalometric x-ray sets, and where quality assurance test object and accompanying analysis software is available, dental CBCT x-ray sets. For CBCT x-ray equipment where QA test object and accompanying software is unavailable, and for hand-held dental x-ray sets, annual testing is recommended.
DD can undertake any the required testing of any of your x-ray equipment.
How long will an X-Ray set inspection take?
Our inspection usually takes 45 minutes for an intra-oral x-ray set (including hand-held x-ray sets) and an hour and a half for a panoramic x-ay set; with a cephalometric attachment some additional time may be required. For CBCT x-ray equipment, the testing may take up to 4 hours.
How much do you charge?
Electro Mechanical Check for intra-oral, panoramic and handheld x-ray equipment, the price of an inspection is £123+vat for members.
For an OPG with a cephalometric attachment the price is £175+vat for members
Routine radiological testing for intra-oral (including hand-held x-ray set), panoramic (without cephalometric attachment) and handheld x-ray equipment, the price of an inspection is £158+vat for members.
Testing for panoramic with a cephalometric attachment the cost is £263 +vat and for CBCT x-ray set the cost is £735 +vat.
What will our engineer do during a routine radiological test?
- Ensure the x-ray set is switched off
- Check serial number of the x-ray tube head
- Mark down the operating potential [kV], tube current [mA] and filtration values from the tube head
- Check the stability of the x-ray mount
- Check for any damage to the x-ray tube head
- Check the electrical and mechanical safety of the machine and provide an assessment of its condition
- Switch on the x-ray set
- Check the condition and operation of the x-ray exposure control
- In the case of an intra-oral x-ray set, check the condition and operation of the counter-balanced arm and fittings
- In the case of a panoramic/CBCT x-ray set, check the rotational movement of the x-ray tube head and the fittings
- In the case of a cephalometric x-ray set which uses direct digital imaging, check the scanning motion of the image receptor with the x-ray tube head and, where applicable, the secondary collimator.
- Check the operation of the dead man switch and/or emergency stops
- Check the operation of the equipment warning lights and audible signals
- For the routine performance testing, complete the required number of exposures and record the measurement results (eg patient entrance dose [mGy], operationating potential [kV], exposure time, filtration)
- Undertake appropriate radiation measurements
- Complete ‘Radiological Inspection’ documentation, accurately completing all relevant fields
- Photograph the x-ray equipment as a record of any concerns noted on your paperwork
- Highlight any discrepancies to you in relation to the electro-mechanical operation of the x-ray equipment
- Ensure that the exposure settings are as found on arrival
- Switch the x-ray set off
- If you have Radiation Protection Consultancy (RPC) contract with DD, the results will be sent to DD's RPA/MPE for review and, for the routine performance testing, a brief formal report will then be issued to the practice